syngo Aortic ValveGuide
- Automated workflow through fully automatic aortic root segmentation, indication of anatomical landmarks and perpendicular view of the annulus – in seconds
- Improved guidance through landmarks and overlay of segmented 3D structures with live 2D fluoroscopy
- Dose and contrast medium savings due to 3D guidance based on syngo DynaCT Cardiac acquisition and ideal automatic C-arm positioning without additional fluoroscopy
How TAVI procedures can be supported by syngo DynaCT Cardiac & syngo Aortic ValveGuide
Step 1: syngo DynaCT Cardiac acquisition
syngo DynaCT Cardiac creates CT-like images of the aortic root and its surrounding structures right in the cath lab or hybrid OR based on a 5-second rotational angiography run with a maximum of 25 ml contrast agent. Selection of the appropriate valve can be based on aortic root diameter measurements combined with TEE.
Step 2: Automated segmentation and landmarking
After the syngo DynaCT Cardiac acquisition and 3D reconstruction the software automatically starts the segmentation of the aortic root. In just seconds the segmentation result is displayed including anatomical landmarks and “perpendicularity ring”.
Step 3: Planning and preparation of valve replacement
The volume-rendered aortic root with its marked coronaries provides a good understanding of the patient’s individual anatomy, which facilitates planning of the procedure. When the 3D volume is turned, the perpendicularity circle degenerates into a straight line if the three lowest cusp points are aligned. This corresponds to an optimal perpendicular angulation for valve implantation. No additional fluoroscopy is needed to find this projection as the C-arm angulation can automatically be synchronized with the 3D view.
Step 4: 3D overlay on live fluoroscopy
By simply activating syngo iPilot on the tableside touchscreen an overlay of the rendered 3D visualization onto fluoroscopic images is created. No additional registration steps are needed as both images are acquired on the same system. Switching to a contour view allows focusing on the essential information. The overlay dynamically adapts to C-arm rotations, zoom and table movements, thus the valve positioning and deployment can be guided very efficiently.
Step 5: Follow-up/Post deployment check
With syngo DynaCT Cardiac a follow-up 3D acquisition to check for correct valve position is possible while still in the cath lab/ hybrid OR right after the valve deployment.